Is the Desktop Kit 3.0 (or a new version) going to be available again in the store? If yes, can ...
Public Lab is an open community which collaboratively develops accessible, open source, Do-It-Yourself technologies for investigating local environmental health and justice issues.
As an open source community, we believe in open licensing of content so that other members of the community can leverage your work legally -- with attribution, of course. By joining the Public Lab site, you agree to release the content you post here under a Creative Commons Attribution Sharealike license, and the hardware designs you post under the CERN Open Hardware License 1.1 (full text). This has the added benefit that others must share their improvements in turn with you.
Hi, I am new to spectometry and was wondering if the spectrometer built with the kit could detect fetanyl in street drugs. In my comp sci class at university this term, we are using the Atmel SAM4S Xplained Pro with which I thought I could build a spectrometer for this purpose (either with or without the kit).
Reply to this comment...
Hi @nyck31, The short answer to your question is no. The spectrometer will measure light in the visible and perhaps short IR wavelengths. Fentayl does not absorb in this region. If you went deep enough into the uv you could see an absorbance, but its intensity should not be very high. As a result, even a uv measurement would not be sensitive enough to make the method useful for the application you envision. Sorry, Jack