The lights I have that are still CFL and not LED are all streaked/wide spectrum lighting with und...
Public Lab is an open community which collaboratively develops accessible, open source, Do-It-Yourself technologies for investigating local environmental health and justice issues.
Public Lab chatroom
Reset your password
SarahC asked on May 31, 2018 05:15
154 views | 3 answers | #16412
The lights I have that are still CFL and not LED are all streaked/wide spectrum lighting with undefined lines.
Look at this perfectly sharp one!
Mine looks like this - even with an eyepiece spectroscope I got from eBay:
As you can see the CFL has smeared blue lines. =(
I'd love to do a "One-click" on Amazon to get a CFL that has the very defined lines of cheap phosphor and bad filtering!
Log in to comment
Hi, what kind of webcam are you using? And what width slit? I think higher resolution may help more than another light, but that's a guess. You spectrum does look good though!
I have an i-phos hobby model!
0.5mm slit, hand held near the light. it's a great little machine. The light though - it's streaking my blue bar...
Oh, hmm, I'm not familiar with that; can you post a picture and/or a link? Thanks!
Is this a question? Click here to post it to the Questions page.
thank for sharing
Both of the examples you link to are over exposed. The over exposure is clipping off the sharp peaks in the spectrum. You can tell if your spectra are over exposed by looking for over saturated colours that appear to be approaching white. Also Spectral Workbench shows where the individual R,G and B outputs are clipping at the top of the display. The simplest way to reduce over exposure is to increase the distance between your Spectrometer and the light source.
Sign up or Login to post an answer to this question.
Standalone Spectrometry software & uploading JPG or PNG spectra
updated 8 days ago
My uploaded file will not process.
updated 16 days ago
How can I get the absorption spectra once I have calibrated my spectrometer data?
updated about 1 month ago
Why do raw RGB values differ from calibrated RGB values? I thought calibration just set pixels to wavelength (nm)?
updated about 2 months ago
Bug : Impossible to calibrate this mercury vapour spectrum
updated 2 months ago
Bug? My Spectrum loads then vanishes.
Has anyone experienced difficulty to log in this site? I having the trouble since yesterday. Not just log in, but to save spectrum too (sometimes can log in but take a long loading). sorry for my bad english.
updated 3 months ago
The spectrum gets vertical no matter how I upload the picture.
updated 4 months ago